[Solved] Problems with conecting to FabScanPi
Can anyone tell me what to do to come past this message?
i'm completely new to raspberrypi so im a bit on ice here.. :-)
there is no erros when the raspberry boots up.. i get ip 192.168.0.51 at boot up, but i cant come thrue to the scanner
i hope someone has the solution
The FabScanPi server is currently not running
If you just started up your Raspberry Pi, please wait a couple of seconds, then try to refresh this page.
If the issue persists, please log into your Raspberry Pi via SSH and check the following:
- Verify that the process is running:
ps -ef | grep -i fabscanpi | grep -i pythonshould show a python process:
pi@fabscanpi:~ $ ps -ef | grep -i fabscanpi | grep -i python
root 14674 1 77 11:19 ? 0002 /usr/bin/python /usr/bin/fabscanpi-serveri
--config=/etc/fabscanpi/default.config.json --settings=/etc/fabscanpi/defaul t.settings.json
- If it isn't, the question is why. Take a look into
/var/log/fabscanpi/fabscanpi.log, there might be an error logged in there that helps to determine what's wrong.
- You might also want to try if you can restart the server (if no obvious error is visible):
sudo service fabscanpi restart.
I think it is the same bug as many people have by using the current version (0.8.0). See https://fabscan.org/community/hardware/hat-not-working-same-as-everyone-else/ for further information. Have a closer look to my latest post in this thread.
I am not at Home at the moment. I will have a woking FabScan available on new years eve. So i am able to test my fix in a working setup. After a a final test in a fully functional FabScanPi, i will release a new image which will include the fix. Until this is not released you will have only one option.
Flash the firmware manually how it is described in the post i mentioned above. Additionally you can patch the FSSerial.py file like it is described in the linked GitHub posting. It is not that hard to do, but you need more advanced Linux skills for editing this File with a text based editor on the terminal console.
Hey Mario... Thank you for the new software, now everything is connecting, and calibrated, i even did my first scanning today, and it looked pretty much okay, considering it is my first scan... looking forward to make a lot of scans :o)